##
455914 Isomotive Dielectrophoresis (isoDEP): Characterization through Particle Velocemitry

The DEP force of a homogeneous spherical particle is *F _{DEP}*=2π

*ε*

*where*

_{m}a^{3}Re[f_{CM}](gradient(E^{2}))*ε*

*is the permittivity of the fluid medium,*

_{m}*a*is the particle radius,

*f*is the Clausius-Mossotti factor,

_{CM}*E*is the electric field. The

*f*governs the magnitude and direction of DEP-induced motion and is a function of particle composition. The imaginary component of

_{CM}*f*can be measured with electrorotation (ROT) [1,2] and the resultant rotational rate is a function of AC frequency as well as the dielectric properties of the media and cell. ROT’s most significant disadvantage is its limited throughput [1]. Further, a common characteristic of existing DEP electrode geometries is that the DEP force is significantly non-uniform in the sample region due to the gradient of the field-squared (

_{CM}*gradient*(

*E*)), making parallel analysis of cells non-trivial.

^{2}The unique characteristic of isoDEP is that the experimental region has a *constant* value of *gradient*(*E ^{2}*) resulting in a uniform DEP force inducing a constant (isomotive) particle translational velocity. Inspired by initial analysis by Herbert Pohl [3], we have developed a modified electrode geometry for isoDEP. Fabrication of extruded electrodes is straightforward via microfabrication methods (DRIE of conductive wafers) or sub-millimeter machining. A sample is injected and flow is halted before field activation. Digital images will extract particle size and, due a constant

*gradient*(

*E*), the only unknown for each particle is

^{2}*f*. The field is applied and

_{CM}*f*is extracted through particle tracking. The particle’s velocity will change as the AC frequency is swept over a specified range to obtain a comprehensive

_{CM}*f*spectrum. Through simultaneous particle tracking such spectra are obtained for every particle in the imaging area, enabling parallel analysis of particles. In the future, isoDEP will be used to extract the dielectric properties of each cell (ex: membrane capacitance) – these properties directly correlate to the cell physiology.

_{CM}References

[1] S.I. Han, Y.D. Joo, and K.H. Han, “An electrorotation technique for measuring the dielectric properties of cells with simultaneous use of negative quadrupolar dielectrophoresis and electrorotation”, Analyst, 138, 1529, (2013).

[2] M. Cristofanilli,, G. De Gasperis, L. Zhang, M.-C. Hung, P.R.C. Gascoyne, and G.N. Hortobagyi, “Automated electrorotation to reveal dielectric variations related to HER-2/neu overexpression in MCF-7 sublines”, Clinical Cancer Research, 8, 615, (2002).

[3] H.A. Pohl, *Dielectrophoresis: The Behavior of Neutral Matter in Nonuniform Electric Fields*, Cambridge; New York: Cambridge University Press (1978).

**Extended Abstract:**File Not Uploaded

See more of this Group/Topical: 2016 Annual Meeting of the AES Electrophoresis Society